09:22:57	 From Angela Marinoni : a unique sensor in the inlet line is enough or you mean one sensor for every instrument?
09:25:23	 From Seba : I suppose we should also measure Relative humidity in the inlet line, just after the nafion dryer.
09:27:43	 From Véronique Riffault / IMT Lille Douai : Not at our site, but we could take a look at our data for the next workshop :)
09:27:52	 From Jakub Ondracek (PACC-ECAC) : YES, the aerosol sample should be dried bellow 40% RH.
09:30:02	 From Adam Kristensson : Do we already report level 1 and 2 for AE33?
09:32:04	 From Véronique Riffault / IMT Lille Douai : Not yet for levels 1 and 2, but they should be recalculated from L0 after an agreement on the "harmonization factor" is reached seems difficult if the MAAP/AE33 ratio is modified by firm/software upgrades!)
09:33:28	 From Jakub Ondracek (PACC-ECAC) : I think if the conditions in the sampling line stay the same RH/p/T then we can have one sensor.
